The continued drive towards cell based assays that better mimic the in vivo environment has led to innovative cell culture systems such as the SciFlow™ 1000 Fluidic Culture System, which utilizes capillary channels to enable physiologically relevant fluid flow across cultured cells. In this application note, we present the ability to accurately image the wells of this novel culture system with the ImageXpress® Micro High-Content Imaging Systems to visualize and quantify cytotoxic exposure kinetics.
The SciFlow 1000 System (Figure 1) is an SBS formatted, 96-well device compatible with plate-readers, imaging systems, and automated liquid handling systems. In order to enable efficient flow of fluids, the system is designed with descending well-heights from left-to-right so the bottom of each subsequent well is 500 microns lower than the previous well in the row (i.e. z-height changes). This allows for cascading flow of fluids across the system without the need for external pumps or connected tubes.
However, this changing z-height across the plate presents a challenge for most imaging systems due to the inability to accurately determine a focal position in each well. This application note will show that the ImageXpress Micro Systems are capable of dynamically adapting focus position across the required 4000 µm vertical range to rapidly deliver clear, focused images for quantification.
- SciFlow 1000 Fluidic Culture System (SciKon Cat. No. AA-1-50)
- HepaRG Cells (Triangle Research Labs Cat. No. NSHPRG)
- Culture media MH100, MHTAP, MHMET (Triangle Research Labs)
- CellTox™ Green (Promega Cat. No. G8731)
- Hoechst DNA Dye (Sigma Cat. No. 33342)
- Tamoxifen (Sigma Cat. No. T5648)
- Fluorescein salt (Sigma Cat. No. F6377)
- Instrument: ImageXpress Micro HighContent Imaging System (Molecular Devices)
Method for optimizing focus position using the ImageXpress Micro system
To determine z-focus position in each well, HepaRG cells were plated in the SciFlow 1000 System then analyzed via nuclear DNA staining with Hoechst. The z-height adjustments for the system are listed in Table 1, with Column 7 being the midsystem or zero-reference point used for configuring laser autofocus during imaging.
Our method workflow (Figure 3) outlines the experimental design (Figure 2) of Tamoxifen treatment of HepaRG cells. HepaRG cells were plated in 50 μL media. Following cell attachment, the media was exchanged with media containing 1:2000 dilution of the cytotoxicity indicator CellTox™Green and 0.5 μg/ml Hoechst (100 μL/well). Then, 400 μL of media containing CellTox Green was applied to the source wells. The plate was incubated for 30 minutes to activate fluid flow and allow for connection of the fluidic channels. Once fully connected, 100 μL of test compound, vehicle control, or 1 μM fluorescein (fluid tracer control) was added to the respective source well. To maintain fluid flow over multiple days, 100 μL was removed manually from wells in column 11, and fresh drug, vehicle, or fluorescein standard was added to the source well three times per day. This resulted in a complete turnover of media over four days while providing a constant infusion of reagent in each treatment group.
Automated Imaging with Z-height focal adjustments to accommodate variable well depths
Hoechst (all cells) and CellTox Green (dead cells) staining was monitored by imaging the entire plate at multiple time points. Plates were read prior to compound addition and at 12 subsequent time points over 4 days. Between imaging runs, environmental control was maintained by keeping the plates in an incubator. Images were analyzed with MetaXpress software using the Live Dead Application Module or Custom Application Module to quantitate cell viability.
The SciFlow 1000 System is configured similar to a 96-well culture format. SciFlow also incorporates z-height steps of 500 microns between neighboring columns to create a maximum z-height change of 4 mm between wells three and 11 (Table 1). Rapid imaging of the whole plate is advantageous as the SciFlow 1000 System has active fluid flow and real-time compartment-to-compartment (well-to-well) signaling as a result of temporal toxicant concentration gradients and parentmetabolite interactions.The ImageXpress Micro System was used in widefield mode and utilized laser autofocus (4X) and, optionally, image-based autofocus (20X) to accomodate for the large z-height changes across the SciFlow 1000 System. The ImageXpress Micro system imaged all 72 wells (1 field of view per well) in 90 seconds at 4X and 5 minutes at 20X. Imaging at 20X included use of imagebased autofocus.
|SciFlow 1000 System z-height adjustments|
|microns||+ 2000||+ 1500||+ 1000||+ 500||0||- 500||- 1000||- 1500||- 2000|
The future of high-content imaging is the capability to accommodate more complex and physiologically relevant culture formats. Imaging of non-standard formats provides flexibility in workflow while producing high-quality images for research needs. The SciFlow 1000 Fluidics Culture System is one such format, which can be used to monitor real-time fluid flow, direct cell culture responses, and dynamic compartment-to-compartment signaling of cellular responses. We are able to show that the ImageXpress Micro systems are compatible with variable z-height format plates such as the SciFlow 1000 System. Automated imaging was achieved across greater than 4mm of z-height change, rapidly generating high quality, reproducible data.